# Flow Rate Sensor Interfacing: How to Measure Liquid with an Arduino

Ever wanted to measure liquid flowing through a pipe or container? Wanted to create a control system based on the water flow rate or quantity? For example, while gardening, to measure the amount of water used to water your plants, to prevent wastage. If yes, then this easy DIY project is for you! Here is some step by step instructions on how to measure water flow rate and quantity using an Arduino flow rate sensor.

### Required Materials

Hardware:

1. Arduino/Arduino Clone or make your own custom Arduino board.
2. Water/Liquid flow meter like a Flow meter.
3. Connecting wires.

Software :

### How Does it Work?

The Arduino flow meter works on the principle of the Hall Effect. According to which, a voltage difference is induced in a conductor transverse to the electric current and the magnetic field perpendicular to it. Here, the Hall Effect is utilized in the flow meter using a small fan/propeller shaped rotor which is placed in the path of the liquid flowing.

The liquid pushes against the fins of the rotor, causing it to rotate. The shaft of the rotor is connected to a Hall Effect sensor. It is an arrangement of a current flowing coil and a magnet connected to the shaft of the rotor, thus a voltage/pulse is induced as this rotor rotates. In this flow meter, for every liter of liquid passing through it per minute, it outputs about 4.5 pulses. This is due to the changing magnetic field caused by the magnet attached to the rotor shaft as seen in the picture below. We measure the number of pulses using an Arduino and then calculate the flow rate in liters per hour (L/hr) using a simple conversion formula explained in step 2.

Flow rate sensor

### Connecting the Arduino to the Flow Rate Sensor

The connections required for this flow rate sensor with respect to the Arduino is very minimal. There are only three wires coming from the flow rate sensor. The 5V Vcc (Red wire), The GND (Black wire), and the signal/pulse (Usually yellow) line. Connect the Vcc and GND of the flow meter to the Arduino’s Vcc and GND. The pulse line of the flow rate sensor is connected to the Arduino’s digital pin 2. The Arduino’s digital pin 2 serves as an external interrupt pin (interrupt pin 0). Now you have hooked up your flow meter to the Arduino!

Arduino flow rate sensor connections

### Uploading the Flow Meter Code to the Arduino and Measuring Water Flow Rate

Upload the Flow Meter Code to your Arduino. The code uses an external interrupt on the Arduino’s digital pin 2. This is used to read the pulses coming from the flow meter. When the Arduino detects the pulse, it immediately triggers the pulseCounter() function. This function then counts the total number of pulses.

In this Arduino flow rate sensor, for every liter of liquid passing through it per minute, it outputs about 4.5 pulses. Dividing the total pulse count obtained by 4.5 will give you the total amount of liquid passing through it in liters per minute. Dividing that by 60 will give you the flow rate in liters per hour, which gives us the total amount or quantity of water/liquid that has passed through it. The sensor is accurate to within 3%.

Here’s a video showing the Arduino flow rate sensor in action!

Use this Arduino flow rate sensor with a solenoid valve to monitor and control the water used. You can use it in your gardening system, or interface it with an LCD display for other applications that require you to measure water flow rate and quantity. Have fun with your new Arduino flow rate sensor!

Arduino Flow Rate Sensor

Recommended Posts
• hohm

Hello
I would like to dose watering with a flow sensor, but I do not know how to do.
I have already done the moisture sensor which detects if the ground and dry and which triggers a relay for the watering pump and I would like to water 100ml each time,
my flow sensor and YF-S401.

• geeth

hi,
Can you kindly help me with an avr studio code for the same.?

• Rick Skinker

I am trying to accurately measure the flow rate of water in gpm from shower heads and sinks. Any ideas?

• Biplab Mahata

​Hi,

I am very new with Arduino and also hobbyist, so i cant improvise codes.

I need to calculate velocity (cm/sec) of the water passing through a squire box, (5cm).
As we know 1cm cube area holds 1ml water; so box with WxHxL 5x5x1=25cm cube area hold 25ml water.

So calculation will be Velocity=((1000ml x Flow rate L/min)/25)/60=Velocity cm/sec.

Example: If water flowing 1 Lt/min ….then…

Velocity=((1000×1)/25)60=0.7cm/sec

If water flowing 8 Lt/min ….then…

Velocity=((1000×8)/25)60=5.3cm/sec

Thanks

• CalGuy

I kind of understand how the sensor you used works but would you be so kind to check this type of sensor out?https://docs.wixstatic.com/ugd/956e06_0a161a52b48a4e27ba3c9e7eb76e13ad.pdf This is the type of flow sensor we use in the landscaping business since it can accommodate for way larger flow rates, the thing is that it only has 2 wires and they do have polarity a black and a red one and they connect to the flow sensor ports on irrigation controllers like the weathertrak pro 3 from hydropoint and also the weathermatic sl4800 and numerous other irrigation controllers; I am trying to read from them but have no idea on how to go about doing that.

Thank you so much for your help and tutorial.

• Yee yang

Hi. How do I put date and time stamp on my reading but will only record when flow is above 0?

• Joseph Funk

After i upload this code, when i open the serial monitor it just shows a bunch of boxes with no actual text. An idea what is going on?

• Ethan

That sounds like you don’t have the same Baud rate set for your serial monitor as the code uses. Look for the line Serial.begin(####). The bottom right of the serial monitor should have that same number selected and say (#### baud). Otherwise the Arduino is sending data to the serial port at a different rate than the port is reading, which is why you’d be getting garbage output (which the serial monitor just replace with black squares).

• Larry Fostano

Can you tell me where the error is here?

Stray \342′ in program

frac = (flowRate – int(flowRate))* 10;

• Kaisar

Hello sir I have a problem on my system I used the same kind of water flow sensor you use and I also have a solenoid valve is it possible to put a limit or set point example if 400ml has pass the solenoid will close the pipe . hope you can help me i’m stuck on this and running out of time Thanks in advance .. sorry for my english not my first language :D

• Gari

We are stuck on a school project and could do with some help. Do you have a programme that we can use so that the water supply is cut off if there is water running through the flow meter for more than 21 minutes? We have a flow meter like the one you have and a relay plus stop valve.
Hope you can help

• Crystal Bekkers

How to change this program to make it work with the NodeMCU board? I tried every pin combination possible but can’t seem to make it work…
Do you by any means have a solution to make this work on the nodeMCU 0.9 board?

• Ganapati

As you shown in video, we have the water flow rate, but how can we calculate how much water consumed?

• Scalde

HI thanks for this info, any ideas on how i can add a second flow meter to this script using an UNO? thanks

• Michael

Hello all, I am trying to get the output of the flowmeter to print out to a tiny printer. I am unsure of where to put the code or even what code I should use. I am new to this stuff and Extremely bad at programming. Any help is appreciated! Happy new year, and thanks!

• Md. Nasimul Haque

Hi Arvind Sanjeev,
This tutorial is great. It will be very useful for me. I have some questions. Kindly solve my issue .
1. Can I use this sensor to detect diesel flow rate?
2. Diesel engine pump takes diesel from tank uses a portion of it and sends the remaining diesel through the overflow pipe. I need to use two sensors to find out the fuel consumption and display it to 16*2 LCD.
I want to use Arduino Mega2560 .
Thank you.
Nasimul haque

• OldWiseMan

Hello Nasimul,

The old wise man says use the power of subtraction to solve your problem.
Subtraction:
Subtraction is a mathematical operation that represents the operation of removing objects from a collection. It is signified by the minus sign. For example, there are 5 − 2 apples—meaning 5 apples with 2 taken away, which is a total of 3 apples. Therefore, 5 − 2 = 3.

• Shri

Hey,
The above hack works. Now i want to post the sensor data using the ESP8266 wifi module to a web server. How can i achieve this, the data should be sent 10 times in a minute. I really want to try this on my own, but the ESP8266 module is tricky and sometimes can dmg the module and the UNO. So i don’t want to take the risk of dmging both.
Can anyone help me this stuff. Thanks in advance.

• Kenneth

Sir, I am using Arduino Uno, flow sensor (YF-S20) and a relay. Need to open/close the relay when the flow sensor reaches a reading of 1 liter. I used your code and it works perfectly with my flow sensor. But now I am trying to play the code to my project using the relay to open/close but i don’t know how to link the code regarding the sensor and the relay. Can you please help me to finish my project. Please send me the right code for my project. Thank you!!!

Try this code put your button on pin 4 and relay on pin 5
/*
Liquid flow rate sensor -DIYhacking.com Arvind Sanjeev Measure the liquid/water flow rate using this code. Connect Vcc
and Gnd of sensor to arduino, and the signal line to arduino digital pin 2.
*/
byte statusLed = 13;
byte sensorInterrupt = 0;
// 0 = digital pin 2
byte sensorPin = 2;
int val = 0;
// put your button on pin 4
int startpin = 4;
// The hall-effect flow sensor outputs approximately 4.5 pulses per second per
// litre/minute of flow.
float calibrationFactor = 4.5;
volatile byte pulseCount;
float flowRate;
unsigned int flowMilliLitres;
unsigned long totalMilliLitres;
unsigned long oldTime;

void setup() {
// Initialize a serial connection for reporting values to the host
Serial.begin(38400);
// Set up the status LED line as an output
pinMode(statusLed, OUTPUT);
// put your relay on pin 5
pinMode(5, OUTPUT);
pinMode(startpin, INPUT);

digitalWrite(statusLed, HIGH);
// We have an active-low LED attached
pinMode(sensorPin, INPUT);
digitalWrite(sensorPin, HIGH);
pulseCount = 0;
flowRate= 0.0;
flowMilliLitres = 0;
totalMilliLitres = 0;
oldTime = 0;
// The Hall-effect sensor is connected to pin 2 which uses interrupt 0.
// Configured to trigger on a FALLING state change (transition from HIGH
// state to LOW state)
attachInterrupt(sensorInterrupt, pulseCounter, FALLING); }
/** * Main program loop */

void loop()
{
// fill button start if relay turn off
digitalWrite(5, val);}
if((millis() – oldTime) > 1000)
// Only process counters once per second
{
// Disable the interrupt while calculating flow rate and sending the value to
// the host
detachInterrupt(sensorInterrupt);
// Because this loop may not complete in exactly 1 second intervals we calculate
// the number of milliseconds that have passed since the last execution and use
// that to scale the output. We also apply the calibrationFactor to scale the output
// based on the number of pulses per second per units of measure (litres/minute in
// this case) coming from the sensor.
flowRate = ((1000.0 / (millis() – oldTime)) * pulseCount) / calibrationFactor;
// Note the time this processing pass was executed. Note that because we’ve
// disabled interrupts the millis() function won’t actually be incrementing right
// at this point, but it will still return the value it was set to just before
// interrupts went away.
oldTime = millis();
// Divide the flow rate in litres/minute by 60 to determine how many litres have
// passed through the sensor in this 1 second interval, then multiply by 1000 to
// convert to millilitres.
flowMilliLitres = (flowRate / 60) * 1000;
// Add the millilitres passed in this second to the cumulative total
totalMilliLitres += flowMilliLitres;
unsigned int frac;
// Print the flow rate for this second in litres / minute
Serial.print(“Flow rate: “);
Serial.print(int(flowRate));
// Print the integer part of the variable
Serial.print(“.”);
// Print the decimal point
// turn on pin 3 if flowrate is greater than 4

// Determine the fractional part. The 10 multiplier gives us 1 decimal place.
frac = (flowRate – int(flowRate)) * 10;
Serial.print(frac, DEC) ;
// Print the fractional part of the variable
Serial.print(“L/min”);
// Print the number of litres flowed in this second
Serial.print(” Current Liquid Flowing: “);
// Output separator
Serial.print(flowMilliLitres);
Serial.print(“mL/Sec”);
// Print the cumulative total of litres flowed since starting
Serial.print(” Output Liquid Quantity: “);
// Output separator
Serial.print(totalMilliLitres);
Serial.println(“mL”);
// check if flow volume is > 1 litres
if (totalMilliLitres > 1000)
{digitalWrite(5,LOW);
// reset volume to zero
totalMilliLitres=0;}
// Reset the pulse counter so we can start incrementing again
pulseCount = 0;
// Enable the interrupt again now that we’ve finished sending output
attachInterrupt(sensorInterrupt, pulseCounter, FALLING);
}
}
/* Insterrupt
Service Routine
*/
void pulseCounter()
{
// Increment the pulse counter
pulseCount++; }

• Kenneth

It works!!! Thank you so much. Is it possible to link a universal coin acceptor or coin slot to this code using multiple coins? I am using CH 923 https://www.sparkfun.com/products/11719 .

1. insert your coin code to get ‘coinsValue’
2. then code the formula dispenseVol=rate*coinsValue where rate is how much you like to sell per cent
example if 1 liter cost 50 cents then divide 1000 ml by 50 so your rate is 20

int rate = 20
int dispenseVol = 0

inside the loop after to get the coinsValue
dispensVol = rate*coinsValue;

3. replace this code:if (totalMilliLitres > 1000)
by this code: if (totalMilliLitres > dispenseVol)

4. reset dispenseVol and coinsValue to zero

happy coding

• parag

hello sir i want ur help

• magdi

Dear All,

Kindly, I want failure rate for different control systems components (sensors, valves, switshes, pumps, etc

• theophilus wiznar

i had a problem to interface water flow meter sensor to arduino in simulation. since i didn’t get water flow sensor library in arduino. please tell how can i do?

• rinku

Sir. I am making a project with water flow senser (hall effect base) and 16*2 lcd
Want to calculate total volume of water flow
But i want use only atmega328pu ic..(programed via arduino uno) ant make new pcb with neccesory componet and thing…
Help me in coding

• Nathan Gant

Thanks, Arvind, for the excellent Arduino sketch. I tested my flow sensor using your software code, and came out with the following 11 data points:

3.5 13158
3.6 12118
3.625 12204
3.73 11626
3.75 11492
3.76 11554
3.8 11188
4 10977
4.04 11546
4.33 10309
4.75 8907
pulses Milliliters

Regression line equation: y=22405-2830.266x

So based on above linear equation, the best K value (pulses per liter) to use in the Arduino sketch for this particular flow sensor would be 4.012. That would give me the closest number for metering, with some built-in mechanical variance from the flow sensor.

• Nathan Gant

Bivariate data from flow sensor #2:

4.0 11824
4.012 10920
4.025 11345
4.06 10411
4.08 11939

Data from this flow sensor appears to be randomly scattered. Also I rejected one data point, 4.0175, because I was getting over 20 liters and on a second run, over 30 liters, that’s far more than the bucket capacity of 11.053 liters. I did notice the signal pin next to the 10k resistor was loose, so next time I’ll start soldering the connections to eliminate that possibility. I don’t know if having a 4th decimal point in the K pulse value 4.0175 made a difference, there may be other factors causing the output to show excess flow.

This demonstates a good reason to test your water flows before moving on to the next step.

Sample size: 5
Mean x (x̄): 4.0354
Mean y (ȳ): 11287.8
Intercept (a): 14420.271184913
Slope (b): -776.24800141568
Regression line equation: y=14420.27-776.248x

So this particular flow sensor would apparently want a factor of about 4.33 to give me an accurate flow rate, a little higher than previous sensor (4.012).

Also, for the stats, I used

http://www.alcula.com/calculators/statistics/linear-regression/

because it was very easy to cut and paste my data set into their site.

• JuN

Bro, do you follow all the coding and sensor type as what the website taught?

• Mike Petterson

When you say “In this arduino flow rate sensor, for every litre of liquid passing through it per minute it outputs about 4.5 pulses”
Are you using the YF-S201 sensor as shown in the pictures?
If yes, then shouldn’t the correct value be 7.5 as stated in the data sheet?

• Agenda Généticien

Hi, i believe that i noticed you visited my site so i came to return the desire?.I’m trying
to to find things to improve my website!I suppose its adequate
to make use of some of your ideas!!

• abhilash

im using the same sensor for my project.im getting the out when i compiled the code on aurdino uno but when i executed the same code on the intel galileo gen 2 im getting the output has 0.can anyone help me hoe to sort out this problem

• Justine Fuentes

What if you wanted to measure alcohol inside of a non-alcoholic beverage? How can we program this?

• avinashjakkuva

How can u connect to raspberry pi which doesn’t have a digital pin how will u connect to it

• Emir Porcil

Sir, I need help! I test the water flow sensor(model: YS-S201) with your program code then I use the faucet as the source of water to fill an empty 6.6 L empty bottle. After filled the bottle I checked the serial monitor then I found out that the output liquid quantity is much higher than water filled in the bottle which is a large amount of error. What can I do to fix this? I used a PVC pipe attached with the sensor and the faucet.

• Jeff Nyika

Hi, so I have 2 flow sensors similar to this and Im trying to use them at the same time, connected to an arduino.
Im not a pro with arduino so wanted to find out if it is possible to use 2 sensors hooked to the same arduino to measure different places?
thanks for a response in advance

• Fong

Brother, did you solve your problem with two sensors? Do you mind to share with me how you complete such as coding and schematic diagram because I am doing the same project with two sensors. I would appreciate if you lend a helping hand.

• Andre

Really nice-work ;)
Have a quastion about the code though, if I wanted to change the time interval what would have to change?

I’m guessing it’s just changing the 1000 (miliseconds) for another value, right?

• Clay

Excellent writeup! I was able to get the code to work on the Uno, but it doesn’t work on my Yun. I know it’s something to do with the PIN mapping. Could you help? The interrupt routine is not getting fired. I have the wiring the exact same.

• Julio

Hi Clay Did you figure it out what happens? Also have issues with yun Instead of using serial i use console to read values over the wifi It just print zero when the yf s201 is working I really dont know what to do

• sylvain

Hi guys,

I have a Yun as well. The mapping for Digital pins to Interrupt pins is different on the Yun. Mine works using Interrupt 1 instead of 0. The only bit of code to change is: byte sensorInterrupt = 1; // 0 = digital pin 2

• Vijayanand

can we connect this flow sensor to allen bradley plc, which has only digital input and output. urgent

• David B

Is there a device that will also show me the direction of the water (e.g. negative voltage?)

• Aaron

Hi Arvind,

I was wandering how you knew that this particular sensor creates 4.5 pulses per litre because I am using a slightly different sensor and to get true values of liquid output from your code, I had to change the calibration factor from 4.5 to 65 (sixty five) by trial and error and was wanting to know if there was a more logical process to modifying your code to suits the specs of the flow sensor I am using?

Thanks

• vishal

hey arvind i want to monitor whether water is flowing or stagnant through a pipe not the quantity of water flowing can i do that using this flow rate sensor if so where can i find the code??

• daniel patterson

I am building an aquarium control unit and have most of the code, now im adding the 2 flow meters and it is plowing my mind any help would be greatfull.

• Shubham chhatre

Arvind sir,
i want to turn on or turn off the flow of water depending upon the sensor value. how we can write program for that condition?

• Daniel patterson

i have 2 meters and it is killing me on getting them into my current code any help would be helpfull

• Miguel

Hi Arvind,

Thank you for the tutorial.

I’m using a different sensor but am getting irregular readings even though I have a constant flow (I’m using air, not water).

I can see that the sensor is giving me a constant square wave but my reading keep varying, about 30% up or down.

This is the sensor:
http://uk.rs-online.com/web/p/flow-sensors-switches-indicators/4468315/?origin=null|fp&cm_sp=featureproducts-_-FeaturedProductsContent-_-4468315

Any suggestions would be greatly appreciated.

Kind regards,

Miguel

• Ana

Hi,
We are doing a similar project and we want to add in an LED indicator light for when a certain amount of water flows through the sensor. Do you have code to do that?

Thanks!

• rebin

also i need this led!
did you find ? can i see your project?

• ANDREW

Is it possible to correct for the inaccuracy of the meter with the code?

• love vegetables

How many flow meters could theoretically be attached to a single Arduino? What considerations would I have to make?
thank you

• horoscopo gratis escorpio

Thanks for every other fantastic article. Where else maay anyone get that kind
of info in such a perfect means of writing? I have a presentation subsequent week,
and I am at the lopok for such information.

• swanand

Hello sir,

can you please provide me the code to interface the sensor with 8051 ucontroller.

• chloe

Hello everyone, can someone give an advice about what is important/advantages to know the flow rate of water?
instead of to gardening, why we must control the flow rate…please, give an opinion..

my dear friend by controlling water flow you can make automatic system for watering the garden, you can fill constant quantity of liquid in bottles/cans/anything, these are some very basic examples , there are many magics happening with water flow rates

• udin

Hello sir,
Im wondering, based on the sensor datasheet, the flow-pulses for 2L/minute= 16 Hz, so, what is the output voltage for 16Hz frequency?

• udin

Hello sir,
Im wondering, based on datasheet
it was flow- pulse for 2L/min=16hz, what is the voltage for 16hz?

• Rahul

Greetings friend.
I created a similar system with flow sensor DN32 and I have very irregular readings is very unstable.
You know how to improve the stability of measurement readings.
I am measuring the fall of flow and for some reason my system is very unstable.
If you know how to improve presicion of readings.

• Craig Larson

Rahul and others above with calibration problems: The datasheet, or something else I read, specified that the meter needed to be installed within 5 degrees of horizontal. Maybe this is the calibration problem? The hack shown here has the meter at 90 degrees to the horizon. I can’t figure out why this would matter since the hall sensor should be indifferent to gravity. And the flow should be indifferent to gravity.

• ratnesh

thank you sir. I am using water flow sensor model YF-S201 having a different calibration factor than yours.What modification is required in the code, also how one can measure rpm of the rotor.

• Erik

Is the totalMilli keeping track of the total amount of water in mL flowed through for just that one measurement cycle or for the entire time the unit is turned on? If I wanted to keep track for the total amount of water flowed through, for an entire day perhaps, how would you suggest this be done?
Many thanks!

• Matt

Hi thanks for the great tutorial. I was wondering if you could explain how one would interface this with a LCD display. And would there be any reason that you couldn’t do a conversion in code to make the displayed value gal instead of liters? Thanks.

• Arvind Sanjeev

Hi Matt,

• Matt

Hi Arvind,

Thanks – your tutorials worked perfectly. I now have a UNO running a flow sensor and displaying results on my 20×4 lcd. Couple more questions if you don’t mind. How could I convert the l/min to gallons? And could you suggest a way to keep a running total of the fluids used? I’m trying to monitor how much rain barrel water we are using annually to water our garden. Sorry for the “entry-level” questions but I am very new to this. Thanks -Matt

• Arvind Sanjeev

1 L = 0.26 Gallon, you would have to convert it using this formula. And to store the total amount of fluid, you would have to save the data in EEPROM. Check for arduino EEPROM tutorials: http://www.arduino.cc/en/Tutorial/EEPROMWrite

• amar mehdi

void open_valve() {

digitalWrite(RELAY_A, HIGH); // turn RELAY_A on
// Serial.println(“Valve Open”);
Serial.write(“{valve_open}”);
}

void close_valve() {
digitalWrite(RELAY_A, LOW); // turn RELAY_A off
// Serial.println(“Vavle Closed”);
Serial.write(“{valve_close}”);
}

sir, this part of code I did’nt underatand ? for what purpose it has been used?

• Arvind Sanjeev

Hello Amar, where did you get this code from, I have not written it? This is my code: http://diyhacking.com/projects/FlowMeterDIY.ino I am assuming that this code turns ON/OFF a solenoid valve to control the flow of water.

• januar

can you help me if i want to display output from sensor to web (chart) by using php

• Arvind Sanjeev

Read the data from the serial port of the computer and have a script which posts that to a webpage. Integrate serial communication and http modules for this, you can use python.

• amar mehdi

sir,
i,m using the same sensor as you use YF-S201. can you show your formula of converting of counts to ml/minute or hour as u did in your video.
because rpm=rpmcoun*60/4.5 does not work correctly.

• Arvind Sanjeev

Please check the source code on the final step of this tutorial.

• amarmehdi

sir, you said that solenoid valves control the flow but you did not use any solenoid valve in your video.
i,m asking about open and close valve function that you use in your code.for what purpose they are used.thanks

• Arvind Sanjeev

Please highlight the code you want me to explain, I dont think I have used any open/close valve commands there.

• Kenneth

Sir, I am using Arduino Uno, flow sensor (YF-S20) and a relay. Need to open/close the relay when the flow sensor reaches a reading of 1 liter. I used your code and it works perfectly with my flow sensor. But now I am trying to play the code to my project using the relay to open/close but i can’t fix it. Can you please help me to finish my project. Thank you!!!

• Chuck Colby

hey Arvind. I have been reading your work and find it interesting for sure. I’m looking to measure the volume of air flow through a tube of air flow traveling at 15 LPM. The goal is to control a pump to allow “x” amount of air through the pipe and then shut off the pump. Why i can calibrate the pump and control the time that the pump runs, then infer the air volume from those numbers, the pumps are not consistent and don’t necessarily run at 15 lpm all the time. Therefore, the ability to actually measure the volume would be ideal.. Do you think your above project would measure air volume?

best,

Chuck

• Arvind Sanjeev

Hi Chuck, Ideally it should. This sensor consists of a simple propeller like system, which turns when water/air hits its blades. Dont know about the accuracy, but the output from the sensor would be definitely proportional to the input air/water. Would suggest you go by with a trial & error approach.

• irfan

sir, i read your code.plz can u elaborate the purpose of valve open and close.i didn,t understand it. my email id is [email protected]

• Arvind Sanjeev

Solenoid valves can control the flow of water based on the sensor data. Can you explain in detail what you want to know?

• Ratnesh

Sir,
Which Arduino you has used kindly tell me how to buy it in india. How you run the software on windows 8 to measure flowrate

• Arvind Sanjeev

• ratnesh

thank you sir. I am using water flow sensor model YF-S201 having a different calibration factor than yours.What modification is required in the code, also how one can measure rpm of the rotor.

• yunus

hye..i want to ask.. i have problems with sensor flow rate.. i want to make the sensor have to dectect flow rate of fluid. if the sensor low than 2ml/min it will close the pump. did you have any idea or any link to refer

• Arvind Sanjeev

You can use a relay to control the pump based on the output of the flow rate sensor (simple if/else statements).

hai, I want to ask about the program. How you program it?

• Arvind Sanjeev

If you are new to arduino, you can check out this free eBook to help you on programming and getting started: http://diyhacking.com/
The code for this project is available in this page itself :)

I am a student from Polytechnic Mukah Sarawak, Malaysia. Our project need this. Now my lecturer want me to build a circuit interface for each one of the sensor. We use 4 sensor, temperature sensor, flow rate sensor, water level sensor and pressure sensor. So, it’s all 4 circuit interface right. If you don’t mind to share any idea. This is my email [email protected], my wechat ID DayangM85.

• Arvind Sanjeev

Hi, you can search online for arduino projects having these sensors. You can then integrate all the code together :)

• Guillaume

I have a problem about the display of the flow, when i run your code there is no information printed on the serial monitor :/ do you know where is the mistake?
I have the Water flow sensor yf-s201 and im connected on the good COM (COM17 for me)
And sorry for my english i’m french ;)

• Arvind Sanjeev

I think there might be some connection problem. Why don’t you try printing something on the serial monitor and check.

• Viswa Devan

Hi Arvind,

Can i know what is your sensor output?
Can i try out the code with using a 4mA-20mA output from sensor?

I’m currently doing a project where i can measure air flow so i can estimate my savings i’ll get by savings air consumed by my tool.

Thank you very much…

• Arvind Sanjeev

Hey, all the info is in the tutorial :)

• Shaun Reed

Hi Arvind,

Your code works great! While I’m measuring flow, I activate my valve, which disrupts the flow measurements. I think this is due to the interrupt of the flowmeter being disrupted. As you suggested above, how do I modify your program to use a digital pin, say pin 4 or 7, instead of pin 2.

Thanks,
Shaun

• Arvind Sanjeev

Hi Shaun,
I fail to understand how the activation of a valve affects the interrupt? Also, why don’t you try the other interrupt pin 1 (digital pin 3).

• Fabio

Hi Arvind, I used your code on a Arduino nano 3.0 and it worked perfectly; after I used the same code on an Intel Galileo: the ide compiles the code but it isn’t able to read to measure from the sensor. Any suggestion?
Thank you very much

• Arvind Sanjeev

Hello Fabio,
Have you used the same sensor on the galileo? And also check the wiring connections for it, and whether the pin number for the arduino program matches that of the galileo.

• Ali

good day Arvind and thanks for this

the displayed out put is ony line of squares, unreadable simples.

what is the port setting you used?

• Arvind Sanjeev

Hello Ali, I used a baud rate of 38400.

• Sukanya

Great Aravind!
Can you please help how to use this code in ARM cortex3.This is little urgent for me.

• Arvind Sanjeev

Hello Sukanya, you would probably have to port the corresponding functions used here. You just need to write a piece of code that reads the digital data from a pin on your controller.

• Ali

Hi Arvind Sanjeev
do I need to attach any thing to the arduino except the wire connection of the sensor and the labtop?

• Arvind Sanjeev

Nothing else, I guess.

• Sean

Hey your code worked great. I appreciate it. If i wanted to output the data to .csv file how could do that? instead of a com monitor.
Thanks for the help

• Arvind Sanjeev

Hello Sean, if you want to output your data to a csv file, I would recommend writing a python script which reads values from the serial port (Arduino) and writes it into a text file in the form of CSV.

• JV

Hi Arvind Sanjeev,
can I use multiple flow meter in one arduino? what will be the configuration of connections? what will be the configuration on the program that you used?

• Arvind Sanjeev

Hello, this program uses the arduino external interrupt pin (digital pin 2) and since there are only 2 interrupts on the arduino, you can use two of these sensors for this program at the same time. However, you can also edit this program to add the desired number of sensors by simply connecting them to the digital pins.

• Droncs46

Could You describe more in detail how to connect 2 or more to a single Board and that the readings were in different variables.

• Arvind Sanjeev

Since this code uses interrupts, you can only add one more to its other interrupt pin (digital 3, arduino uno has only two interrupt pins). However, you can change the code to have other pins as interrupts too. Check this: http://www.geertlangereis.nl/Electronics/Pin_Change_Interrupts/PinChange_en.html

• Droncs46

BOARD int.0 int.1 int.2 int.3 int.4 int.5
UNO, Ethernet 2 3

Mega2560 2 3 21 20 19 18

Leonardo 3 2 0 1 7

• Droncs46
• Ashok

hi Arvind sanjeev,

• Donald

Hey, thanks for the video! I am using the adafruit sensor, but when I used your code, there is no printed information on the screen. Do I need to change some settings?
Thanks!

• Arvind Sanjeev

Hello Donald, can you please share the link to your sensor? I am using another China made sensor for measuring the flow rate.

• Faiz Wahyudi

• Syazwan

Good days sir.. I try to upload the sketch to my arduino UNO and run it. I can’t find the result as what you get sir. Im using G1/4 Flow Rate sensor.. Can you help me..

• Arvind Sanjeev

What is the output being displayed on the serial monitor?